The Tracker Heart Rate Monitor does not appear to be giving consistent readings, although the wireless connection seems strong.
In some cases it has been demonstrated that a very close proximity between the I.T. and the Polar heart rate strap can cause erratic readings to be displayed in Tracker Version 5.
Consider re-attaching the heart rate I.T. to the far end of the chest strap, away from the heart rate monitor.
While perusing our support articles, if you find an article confusing or you feel it could be written better, just click add a comment and leave your concerns.
Our Customer Service department will attempt to address your concerns as soon as possible. Thank you for helping JTECH help you☺